• X-ray Generator: Stationary anode, double focus, and bridge Silicon rectification.
The X-ray generator is an oil-immersed self-cooled unit.
■ Max rated capacity: Current: 15mA/30mA/60mA/100mA
■ Power supply: Voltage: 180-240V; Frequency: 50Hz Power: not lowerthan 7kVA;
Résistance: not larger than 10
■ Voltage adjustment range of X-ray tube: 50~90kVp
■ Range of the timer: 0.08-6.3s
• Distance between the focus to the ground: 700-1780mm
• X-ray tube focus transverse rotation: ±60°
• Spécification of X-ray tube: Model XD4 2.9/100 fixed anode, double focus
• Transport dimension (LxWxH) (mm): 1230x900x2050
■ Weight (kg): Net: 159 / Gross: 256