We are providing the highest functional pneumatic tourniquet systems in market. It provides suitable functions for an operating room under the optimized conditions. (Recording, LOP sensing, LCD Touch Screen, Safety Alarm System, Battery mode).
When Alarm sound generates
At the end time, alarm sound generates to ask to stop the surgical operation by pressing “DEFLATE” touch key or by pressing “TIME” touch key to increase operation time. Even though turning off the alarm sound by alarm button, it generates by 30seconds continually until “DEFLATE” or “TIME” button is set.
Leakage error is indicated on the screen and the alarm sound generates, the error can be solved by changing cuff after pressing the alarm button.
No cuff error is indicated on the screen and the alarm sound generates, the error can be solved by connecting a hose or a cuff.
How to use LOP sensor
For the suitable LOP sensor use, please set the LOP sensor to the thumb of arm to be operated or the toe to be operated. The machine checks the limb occlusion pressure for 15seconds and set automatically the optimized pressure by adding safety default pressure 75mmHg.
It records the time, pressure and error used during operation with real time clock.