Universal cannula III for adultsThe universal cannula III is particularly quiet.In a study by the Fraunhofer Institute, the measured volume was significantly below the level of comparable competitor products. It provides tested, low-noise aspiration of spray mist. And that without having to leave the pressure points. Proven reduction of the germ-laden aerosol cloud 30% lower noise level during suction – confirmed by expert assessment No corners or sharp edges – comfortable in the patient's mouth Stylish colours, to match every surgery interior
A risk analysis and categorisation of medical
products often used in dentistry must be performed before their reprocessing by the operator.
Comply with all national directives, standards and
specifications such as e. g. the "Recommendations from the Commission for Hospital Hygiene
and Infection Prevention".
Accessories of the medical device are also subject to reprocessing.
Classification recommendation given intended
use of the product: semi-critical B to critical B
Semi-critical medical product:
A medical product which comes into contact
with mucous membrane or pathologically
Critical medical product:
a medical product which also comes into contact
with injured skin and blood.
The operator is responsible for correct classification of the medical products, defining the reprocessing steps and performing the reprocessing.
The reprocessing procedure was validated as follows:
– FD 350 disinfection wipes (Dürr Dental)
– Manual cleaning
– ID 215 enzymatic instrument cleaner (Dürr
– Cleaning brush
– ID 213 Instrument disinfection (Dürr Dental)