The outersurface of the device has been produced of galvanized sheet with electrostaticpaint resistant to rust and the inner surface is produced of stainless steelsheet.
Blood bank refrigerator operates with 195-230 V/50 Hz network voltage.
The door ofthe device has double thermal glazing, it can be locked and has magnetic sealrings. Thus, it provides an opportunity to trace the stocks.
• Each shelfin the refrigerator takes 12, 30, 40 and 56 pieces of 450 ml whole blood bagsaccording to the refrigerator capacity.
The devicehas a pool type stainless steel shelf system. There is a plexi-glass protectorin front of each shelf. This mobile shelf system can be easily moved.
There areseparators inside the drawers helping the blood bags to be placed and stockedeasily.
Blood bank refrigerator operates between +2/+8°C, which is the ideal storage temperature of bloodingredients, and it is set to +4°C.
Blood bank refrigerator provides equal temperature distribution with the strengthened fan system itcontains.
Thelightening of the inner cabin of the device is carried out with LED lights.
Blood bank refrigerator has full automatic defrost system in order to protect the productivity of thecooling evaporator.
Auser-friendly digital control panel with a micro processor has been used on thedevice. This system can keep the data in memory for 30 days.
There is aUSB socket to transfer the cabinet temperature information to the computer whenrequested. In this way, the 10 year old temperature records can be transferredto the PC in excel format.
Thethermostat at the control panel of the device can make measurements with 0.1degree sensitivity.