The irradiator is intended for air disinfection of the premises of healthcare facilities under the conditions of constant ultraviolet radiation to prevent bacterial contamination. CAUTION! The irradiator is not intended for sterilization and disinfection of instruments! The irradiator provides for healthcare facilities readiness for operation in accordance with the standards and requirements regulated by the State Sanitary and Epidemiological Surveillance authorities of the Ministry of Health of the Russian Federation. The irradiator is placed in surgery, dressing, examination rooms, dental offices and other rooms, where work with sterile medical instruments and environment is required, including in category I rooms (germicidal efficiency level of at least 99.9%) in accordance with "R 3.5.1904-04.3.5. Disinfectology. Use of ultra-violet bactericidal radiation for the disinfection of air and surfaces in inner premises. Guidance" (approved by Chief State Sanitary Doctor of the Russian Federation of 04.03.2004).
MAIN PERFORMANCE VALUES
The irradiator operates from AC voltage 220 ±22 V, frequency 50 Hz.
The power consumed by the irradiator from AC mains is no more than 200 VA.
Exposure rate from UV radiation source (two 30W germicidal lamps) at a distance of 1 meter at a wavelength in the range of 253.7 nm is no less than 2 W/m².
The irradiator continuous operation is no more than 24 hours.
Time to recover the irradiator operation mode shall be no more than 10 minutes.
As for safety requirements, the irradiator is class I type B product in accordance with GOST R 50267.0-92
The irradiator overall dimensions are 1200х220х80 mm.
The irradiator weight is no more than 5 kg.