Water distilling apparatus АЭ-14-"Я-ФП"-01 is used as desktop version. The water distilling apparatus is easy to control, the control unit contains the only one "On/Off" button starting the water heating mode in the evaporator. The control unit has indicators with the following functional purpose:
"GREEN" - the control unit power supply voltage is on.
"RED" - the water level in the evaporator is below the permissible level.
"YELLOW" - heaters mounted on the evaporator bottom operate.
The low water level indicator in the evaporator is duplicated by sound piezoelectric element.
The power supply unit of АЭ-14-"Я-ФП"-01 model is connected to AC mains with voltage of 220 V through РШ40-ВШ40/220 V connector (included) and automatic switch (included) to the electrical knife-switch. The cross-section of the lead-in copper wire shall be no less than 4 mm².
The water distilling apparatus requires simultaneous connection to the feed water source and to the waste water discharge point, as it is a flow-type device. Connection is carried out using the hoses included in the set.
The distillate production mode is set manually after the device reaching the operating mode using water feed control tap. It takes no more than 15 minutes for АЭ-14-"Я-ФП"-01 to reach the distillate production mode.
The quality of water produced meets the requirements of Article of FS 42-2619 "Purified Water" of the State Pharmacopoeia of XI edition. At that, the feed water quality shall comply with the requirements of GOST R 51232.
The water distilling apparatus has the additional function of generating boiled water, which can be taken from the evaporator drain tap.