The intensive care hospital set for dialysis (UBS 2220) provides electrical power, low power and medical gasses outlets, and also an adequate illumination.
For the dialyze process, per one patient place, the following is provided:
- three monophase grounded outlets 220V/16A
- one grounded outlet 220V for the dialyze machine
- four potential equalizing connectors (terminals)
MEDICAL GASSES OUTLETS
They are mounted in the separate channel for gasses distribution, as an option. Please specify the needed outlets at ordering.
The integral part of the system is always instrument supporting bar.
The DIN 5035 part 3 recommends illumination conditions in the dialysis rooms.The recommendations are similar as with classic care rooms (general lighting:100 lx, reading lighting:200 lx, surveillance lighting: 5 lx),but the working lighting recommendation is raised to the 500 lx.So intensified illumination is needed at both the beginning and the end of dialyze process, while for the greater part of this process, the general lighting is quite sufficient.
At UBS 2220 hospital set,working lighting with parabolic glossy raster and rastered Plexiglas screen is positioned on the ceiling.
Usually,patient beds are separated with the paravanes in the dialysis room, and may be separated from the walls,so we distinguish two types of the UBS 2220 set:
-for the ceiling montage (UBS 2222)
-for the wall montage (UBS 2221)