vaginal pelvic muscle probe
Liberty Loop Vaginal Electrode
Use Liberty Loop with a Tenscare Pelvic Floor Exerciser to treat stress, mixed or urge incontinence.
Smooth, comfortable, shape
Hollow shape helps probe to stay in place.
Biofeedback pointer to aid correct voluntary pelvic floor exercises
Easy to clean
You can attach the pointer to the base of the Liberty Loop to show the effect of correct muscle contraction. You can use it with stimulation or to aid voluntary kegel exercises.
During a correct contraction the pointer should move downwards (away from you).
The Liberty electrode is designed for single patient use. Before first use, and after each subsequent use, the unit should be cleaned as follows:
Wash electrode including cable in mild soapy water. Rinse thoroughly in clean water.
Dry the electrode with a clean cloth/paper towel and allow to air dry before returning to the plastic bag.
DO NOT immerse in liquid. DO NOT use boiling water.
Clean thoroughly and dispose of according to local regulations
Material: Electrode body: ABS
Electrodes: Stainless steel #301
Cables: PVC over copper tinsel conductor
Weight: 20 gm
Length:73 mm (inserted part)
Width: 34mm dia. (widest point)
Connections: 2mm pins
Electrode surface area: 4.30cm2
"In Use" product lifetime: 6 months
Indications for Use
For use as an intravaginal probe in the treatment of female urinary incontinence with muscle stimulation and EMG biofeedback equipment approved to EN60601-01.