1) Test both IgM and IgG to improve sensitivity
2) Result shows in only 10 mins at the soonest
3) Sample types: finger-pirck blood/whole blood/serum/plasma
4) It also could test the CPR,SAA,PCT,CK-MB,cTnl,D-dimmer,MYO,NT-proBNP through the Immunofluorescence Analyzer.
4) Test method for IgM and IgG :
1.Read the manual and the instruction of applicable instrument carefully before using and selecting corresponding sample type and test item on the instrument.
2.Test card and sample should reach to room temperature before test.
3.Insert ID card corresponding to batch number into the card slot. Carefully check reagent name and batch number displayed in the screen after the card reading.
4.After reagent information confirmed, unpack foil bag and place the test card on the flat surface.
5.Deliver sample (30μl of serum, plasma or whole blood sample) into the centrifugal tube with sample diluent in, mix well and respectively add 100μl diluted sample into the sample port of the test card of COVID-19IgM and COVID-19IgG.
6.The reaction time of both test cards is 10 minutes.
7.Method one: Select Instant Test Mode, insert the test card after the reaction for 10 minutes into the Immunofluorescence analyzer. Test result will be printed by the analyzer.
Method two: Select Standard Test mode, insert the test card immediately into Immunofluorescence analyzer after adding diluted sample. The analyzer will print test result after the countdown of reaction time according to different test items.
8.The final result should be determined by the combination of IgM test result and IgG test result.
9.Calibration: Insert ID card to read the calibration curve.