Antimicrobial Additive Ensures Protection
The alarm fascia incorporates an antimicrobial additive to minimise cross infection.
Area Alarm - Medipoint 26
The alarm is designed to monitor up to 6 gas services in the normal, high and low pressure conditions, and are suitable for installation into any location within a hospital. Alarm panels are usually installed downstream of the area valve service units to provide local warnings of supply pressure/vacuum outside of the normal operating range. Line contact monitoring circuits constantly monitor the integrity of the pressure sensors and interconnecting wiring.
Central Alarm - Medipoint 125
The Medipoint 125 can monitor up to 5 gas services or twenty point alarm conditions and are suitable for installation into any location within a hospital. Each alarm panel may be used as an input or repeater panel, as part of the total central alarm system. Line contact monitoring circuits constantly monitor the integrity of the pressure sensors and interconnecting wiring. In the event of any fault, the line contact monitoring circuits initiate the specific gas service failure indication, a 'System Alarm' indication and the audible warning.
Medical Gas Area Alarm
Each medical gas area alarm panel shall be capable of monitoring
6 medical gas services by means of pressure sensors, which detect
deviations from the normal operating limits of either pressure or
medical vacuum. The medical gas area alarm shall fully comply with
the requirements of HTM2022, HTM 02-01, C11, BS EN 60601-1 and
BS EN 60601-1-2 and BS EN ISO 7396-1
The cover, backbox and bezel (if required) shall be polyester powder
coated in a RAL9010 30% gloss finish.