Delta P can supply alarm systems manufactured in accordance with HTM2022/02-01 memorandum.
In particular the following models are available:
Area alarm panel HTM 2022/02-01
Plant alarm system HTM 2022/02-01
Area alarm panel HTM 2022/02-01 General description
The area alarm panel is used to monitor pipeline pressure within a ward, theatre etc.
The pressure is monitored by pressure switches in the pipeline downstream from the last AVSU. Pressure is monitored for both high and low pressure (low only for vacuum
Plant alarm system HTM 2022/02-01 General description
The Plant Alarm System is intended to be used as central indicator panel in order to display warning and alarm conditions for medical gas supply systems. It must be located in a position where there is continuous 24-hour occupation, such as the telephone switchboard room or the porter’s lodge.
AREA ALARM PANEL HTM 2022/02-01
It allows the monitoring of both high and low pressure (low only for vacuum) up to 6 gases contemporary.
Alarm system fault in accordance to HTM 2022 / 02-01 requirements
Possibility to check of integrity of the cabling between alarm system and pressure switches with a termination box supplied on reques
Designed to meet the requirements of:
HTM 02-01 / 2022 Medical Gas Pipeline Systems
C11 NHS Model Engineering Specification – Medical Gases
BS EN ISO 7396-1 Pipeline systems for compressed medical gases and vacuum
2004/108/EC The EMC Directive
2006/95/EC The Low Voltage Directive
PLANT ALARM SYSTEM HTM 2022/02-01
It allows the monitoring of the status of the supply unit from 1 to 6 gases contemporary, giving up to 4 alarm conditions plus normal for each gas.